Testing Your Test Strips
Test strips have a plastic or paper base with very thin metal or carbon overlay and an enzyme. The blood or control solution runs up the strip and mixes with the enzyme. The meter then passes a small electrical charge through the mixture. The resistance is what determines your reading. Meters are calibrated during manufacturing to equate a blood glucose result equivalent to the level of resistance that would occur in blood with that much glucose present.
The enzyme on the strip is very sensitive to heat, cold, moisture and direct sunlight. Performing a test and comparing your control solution result to the range on the side of a newly opened container of test strips tells you whether the batch will work correctly. Statistical analysis has shown that if one strip from a bottle is good then all the strips are good, and conversely, if one is bad then they’re likely all bad. So, if your test strips were damaged by environmental factors during shipping, you will know instantly that results will be inaccurate and you need a new batch. You should perform a control test the first time you open a new vial of strips and whenever you get a result that doesn’t seem right. If you get a result that’s outside the range, perform another control test right away.
Note: If you get your test strips from Diathrive and find the control solution test shows your strips are not reading accurately, contact Diathrive Customer Service so we can help troubleshoot or replace your faulty test strips.
On Call Chosen Control Solution
For use with the On Call® Chosen and GK Dual meter.
The Control Solution verifies the proper operation of your glucose meter by checking the test strips and meter against a pre-calibrated control solution.
Three levels of control solution are available labeled Control Solution 0, 1 and 2. Control Solution 1 works for almost all self-testing needs. If you think your meter or strips may not be working correctly, you may also want to do a level 0 or level 2 test.
- Free shipping from 60
- Ordered before 18:00, sent same day.
- Questions? Customer service: +31 85 1303 646
What Is The Shell Life Of The Glucose Control Solution
As per manufacturers, the glucose control solutions last for 90 days after the first use. Nevertheless, many manufacturers follow some practices to make the glucose control solutions last longer for like 4-5 months.
Whenever you purchase the bottle of the control solution, check the expiry date. The expiry date should be a minimum of 90 days from the date you purchase. Never use the expired glucose control solution.
After each use, re-cap the control solution bottle tightly. You need to ensure the bottles cap isnt loose otherwise, the air will affect the quality of the control solution.
Dont expose the control solution bottle directly to the sunlight or humidity. The storage conditions are already mentioned on the control solution bottle please follow them.
Don’t Miss: Average A1c For A Diabetic
When To Use Control Solution
Although it is recommended to use control solution weekly, there other times it is a good idea to use it.
- When test results do not match how you feel
- When the meter is new
- When your meter has been dropped or potentially damaged
- When test strips have been exposed to extreme temperatures or humidity
- When multiple tests in a row do not match
- When your meter has been cleaned or repaired
- When your health care specialist recommends a glucose meter calibration
- When using new test strips
- When test strips have been exposed to air for prolonged periods of time
How Often Should The Control Solution Be Used To Test Glucose Level With Glucose Monitor
It is important to run periodic control tests to ensure that the meter and test strips are functioning and giving reliable results. These tests are strongly recommended:
- When the meter is new
- before using a new batch of test strips
- When the cap has been left off of the strip vial, allowing the strips to oxidize
- When the strips have been stored in an area of extreme temperature and/or humidity
- When the machine has just been thoroughly cleaned
- When test results do not coincide with how you feel
- When the test is repeated and the levels are not what is expected
- When you want to verify that you are performing the test correctly
Also Check: Diet For Diabetics With Heart Disease
What Is Glucose Control Solution
Control solution can be explained as “fake blood” that contains a known amount of glucose. It is a safe and reliable method to test the accuracy of glucose meters and test strips. These glucose control solutiona are highly recommended to maintain the health and well-being of diabetes patients. This glucose liquid is put on the test strip just like a normal blood sample and result is recorded to check whether the test strip and meter are working properly. Health Products For You offers a wide range of glucose control solutions from various top-selling manufacturers like Siemens Healthcare, US Diagnostics, Agamatrix, Roche Diagnostics, etc.
How To Use Control Solution
Control solution can be used to confirm that your meter and strips are working correctly. You can purchase control solution for your One Drop meter in our online store. One Drop control solution comes in both Level 2 and Level 4. You may receive one or the other depending on which one is in stock at our fulfillment centers. Either one will give you an accurate measurement.If you are enrolled in One Drop through your employer, please email with your name, email, and employer name to request control solution.
Can I use other control solutions?The One Drop glucose meter must be tested with Agamatrix control solution. Control solutions from other manufacturers may not produce an accurate test result.How to use One Drop control solution:
Determine if you have Level 2 or Level 4 control solution. The level is listed on the box.
Find the reading range that matches your control solution level. The range is listed on your One Drop test strip vial.
Use the control solution to take a glucose reading as you normally would, substituting a drop of control solution for the drop of blood.
What is the difference between Level 2 and Level 4 control solution?
Also Check: Diet Plan For Diabetes Cancer Patient
Control Solution = Better Control
Monitoring blood glucose at home is second nature to many people with diabetes. In fact, its often so automatic that you may not stop to think about all the steps in the process that may interfere with achieving an accurate blood glucose reading. First, theres the near-automatic loss of accuracy that comes from taking a blood sample from capillaries, which is what happens with both fingertip and alternate-site testing. The blood glucose level in your capillaries is delayed compared to the blood glucose level in your veins, which is considered your true blood glucose level and is what lab tests measure. The two levels may differ by as much as 10% to 15%, depending on how quickly your blood glucose level is changing. If youve been fasting and your blood glucose level is stable, the two probably wont differ by very much.
Whats your experience with control solution has your doctor or another member of your health-care team recommended that you use it? If so, how often do you use it? Have you found it to be useful in spotting test strip or meter errors, or has it just confirmed that none of your equipment is malfunctioning drastically? Would you use control solution to check your meter if the solution could be guaranteed to contain an exact glucose level? Has control solution given you more confidence in your meters readings? Leave a comment below!
Best Selling Diabetes Test Strips
What can you say about control solution? Does it work? Yes. It works. Its easy to use. It reassured me that the strips and meter are accurate. The Freestyle products are superb. So good that I buy them even though my carrier doesnt cover them. And an endocrinologist whom I trust recommended them as first rate. Also, dealing with Diabetic Warehouse is fast and easy.
Great shipping time and product was useful with my glucose tester.
Shipping AND Delivery
At Diabetic Warehouse, we’re committed to giving you the most for your money. That’s why EVERYTHING you buy ships FREE via our fast 3-5 day shipping method! Once shipped, you will receive an email with your tracking number. Unlike many other online stores, here at Diabetic Warehouse we stock 90% of what we sell. This enables us to get your order shipped the same day as long as it’s placed before 3pm EST, for prompt delivery.
We offer affordable diabetic supplies at up to 60% below retail pricesespecially for people who are uninsured, under-insured or must pay out of pocket. Save on diabetes test strips, glucose meters, lancets and more from trusted brands such as FreeStyle, OneTouch Ultra, Accu-Chek, Bayer, Prodigy, TRUEtest & many more. Every item we offer on our site qualifies for FREE Shipping with our fast 3-5 day delivery time.
Don’t Miss: Diabetic Desserts Without Artificial Sweeteners
Glucose Control Solution: What Is It
Glucose control solution is a liquid that combines water, glucose, and two key elements called buffers and microbicides.
The point of creating a glucose control solution is to help you check that your test strips and blood glucose meter are functioning correctly.
Glucose control solution is designed to mimic the physiological environment of your blood, and contains a known glucose concentration.
Testing your blood glucose meter using this solution helps your ensure that your devices are calibrated properly to give you accurate data.
Fora God Glucose Normal Control Solution
Couldn’t load pickup availability
It is very rare for a meter to be out of calibration. Control solutions are a way to test your meter so you know that it is giving you accurate results.
FORA GOD Control Solutions contain a known amount of glucose at points relevant to normal, low or upper limits of the clinical range that can react with FORA GOD test strips.
You should perform a control solution test if you experience one or more of the following conditions:
- You suspect the meter or test strips are not working properly.
- You have dropped or think you may have damaged the meter.
- Your blood Glucose test results are not consistent with how you feel, or if you think the results are not accurate.
You can check the performance of the meter and test strips by comparing the control solution results with the range printed on the label of test strip vial. Checking these regularly can ensure your test results are accurate.
Please use GOD Normal Control Solution with following devices:
- FORA Premium V10 / Premium V12
- FORA V10 / V12 / V20 / V30
- FORA D10 / D20 / D30 / D40
- FORA G20 / G30 / G31
Important: The control range may change with each new pack of test strips. Always reference the control range on the label of your current test strips vial.
You May Like: Can You Lose Weight If You Have Diabetes
What Is Control Solution For A Glucose Meter
Glucose control solution is a liquid made up of water, glucose, buffers, and microbicides. These ingredients are vital to creating a solution with the following properties:
- Has a concentration of glucose that can be used to test the functionality of your meter and test strips.
- Has a pH level that is close to the physiological value of human blood.
- Prevents the growth of micro bacteria that would otherwise alter the glucose concentration.
Your blood glucose monitoring system can detect the amount of glucose present in control solution in a similar way to how it detects the amount of glucose present in blood. A glucose control solution reading on your meter appears as a numerical value. This value should fall within the target range printed on the test strip vial, which is based on the concentration of glucose in the control solution.
Manufacturers often have a few options of control solution that are compatible with each brand of test strips. The different options, or levels Level 1, Level 2, or Level 4 for example have glucose ranges that are different from one another, typically corresponding with low, normal, or high glucose concentrations. Your test strip vial should show the target range for each different level of control solution. Be sure to identify the correct range that corresponds to the type of control solution you have.
Glucose Control Solution Expiration
A glucose control solution has two expiration dates. The first is the closed bottle expiration date, which is the last day by which you should open your glucose solution.
The second is the open bottle expiration date, which is three months after you first open the bottle.
Its important to make sure that your glucose control solution isnt expired, as after the expiration date its strength can change.
This will skew the calibration results, which would be counterproductive to the whole point of using a glucose control solution.
In summary, even though it might seem like a hassle, the benefits of testing your glucose control strips and devices can be the difference between great glucose control and a string of glucose difficulties.
Amla Green has strict guidelines for scientific references in our articles, and we rely on peer-reviewed studies, academic research institutions, governmental organizations, and reputable medical organizations. We do our best to avoid using non evidence-based references in all articles. The references in this article are listed below.
Heinemann, Lutz. Control Solutions for Blood Glucose Meters. Journal of Diabetes Science and Technology 9, no. 4 : 72324. https://doi.org/10.1177/1932296815587602″
How to Prepare a Glucose Solution | Sciencing. https://sciencing.com/prepare-glucose-solution-6966226.html”
Don’t Miss: Signs To Know If You Have Diabetes
What Does Control Solution Contain
Control solution is a mixture of sugar, water and other stabilizers. Control solution does not contain any blood cells, interferents, or blood products and hence is less viscous than blood. Control Solution reacts differently than blood when applied to the test strip. Therefore, the results of a glucose control test do not reflect a true blood glucose value.
How Do I Properly Perform A Control Solution Test With Contour Meter
Control Solution Testing: When you open a new vial of test strips If you leave the test strip vial open for an extended period of time If you think your meter may not be working properly If your test results do not match how you feel Low, Normal and High control solutions are available. Use only CONTOUR® control solutions to practice on the system or to run a check to make sure the meter and the test strips are working properly. Using anything other than CONTOUR® control solution could present incorrect results. Remove a test strip from the bottle. Tightly close the bottle lid immediately after you have removed the test strip. NOTE: Check the expiration and discard dates. Make sure the test strip does not appear torn or damaged. IMPORTANT: To ensure accuracy of your blood glucose test results and accuracy of your control results, If your control test is out of the range listed on the bottle,Continue reading > >
You May Like: Is Canned Tuna Good For Diabetics
How To Use Glucose Control Solution
Using glucose control solution is simple, which is why it can be such an easy way to ensure you have accurate control of your blood glucose.
Step 1: Insert a new testing strip into your blood glucose meter, and check that the device is powered on, functioning, and ready for a blood glucose test.
Step 2: Shake the bottle of glucose control solution well
Step 3: Squeeze out the first drop from the glucose control solution bottle and discard it. Then wipe off the tip of the bottle to make sure its clean.
Step 4: Dispense the second drop of control solution onto a hard, clean surface, then bring the test strip up to the droplet until youve collected enough solution. This should be almost the exact same process as using a lancet and collecting a drop of blood for a blood glucose test.
Step 5: Check the test results on your meter or monitoring device, just like you would with a blood sample. The bottle of glucose control solution should provide a range of acceptable values that should come up.
Note: If the results of the control solution test are outside of the range expected, we recommend repeating the first 5 steps once more. If the results are inaccurate again, its worth looking into a new pack of testing strips, or a new monitoring device.
Step 6: Dispose of any used test strips and store the glucose solution in a safe place at room temperature.
How To Use Control Solution For Glucose Meter
The easiest way to ensure whether the blood glucose monitoring system is working appropriately or not is by using the glucose control solution. You can check the blood glucose monitoring system at home by using this control solution. The glucose control solution is the perfect alternative to the finger stick.
The testing of the control solution is the same as checking the blood samples. Nevertheless, the properties of the control solution are different, and the results are used for multiple purposes. Here, the manufacturer decides the amount of the control solution of the particular glucometer.
In a nutshell, the glucose control solution checks whether the test strips and glucometer reads the samples correctly and delivers accurate results.
But, how to use control solution for glucose meter? In todays post, well explain the control solution for the glucose meter and how to use it for testing purposes. Lets get started.
Read Also: Weight Loss Supplements For Diabetics